All Home Care Monitor in our shop are 100% brand new, workable and checked carefully before shipping out. Order it now with confidence!
1. Handheld network type, small and light, easy to atomize anytime and anywhere. Independent atomization components, one-click disassembly, easy to clean and disinfect, avoid cross infection.
2. The fog volume is large, the atomization is fine, the particle size is ≤3.7μm, and the drug absorption is more sufficient. Microporous spray tablets, atomization speed ≥0.25ml/min, the residual amount of liquid medicine is less than or equal to 0.5ml, economical use.
3. Through the piezoelectric element, the low-noise design is quieter, lighting means normal operation, flashing means no battery or insufficient voltage, when there is no medicine in the cup or the voltage is insufficient, it will automatically shut down. 2 AA alkaline batteries or use a USB power cord.
Product Name: Grid atomizer
Liquid medicine residue: ≤0.5mL
Atomization speed: ≥0.25ml/min
Atomized particles: about ≤3.7μm
Medicine cup capacity: minimum about 0.5mL maximum 10mL
Noise: ≤50dB (A)
Working frequency: 110KHz±10%
Power consumption: 2.0W
Power configuration: DC5V/1000mA USB power cable or 2*1.5V (2 section No. 5 AA alkaline, the product does not include batteries)
1. Remove the top cover, take the cup and silicon gasket out.
2. Add 30ml purified water into the Water Chamber, the water level should be 3-5mm above the external silicon ring.
3. Fill the medicine cup half-full. put the cup into the chamber.
4. Re-install the top cover securely with the silicon gasket.
(Note: the best atomisation temperature in the chamber is 30℃/8℉)
Please use saline water or WARM(NOT boiling) water to sterilize the accessories.
1. Do not rinse the cup directly with running water
2. Clean the surface of the medicine cup, the medicine cup can be removed separately, and the main machine can not be washed with water.
3. You can’t touch the spray net of the spray port with sharp things such as cotton swabs and toothpicks.